ActiGraph's Response to Novel Coronavirus (COVID-19) - June 25th Update
ActiGraph employees are working remotely until July 13th, 2020. Shipping delays should be expected during this time.
24 Hours of Sleep, Sedentary Behavior, and Physical Activity with Nine Wearable Devices.
- Published on Oct. 17, 2015
Background: Getting enough sleep, exercising and limiting sedentary activities can greatly contribute to disease prevention and overall health and longevity. Measuring the full 24-hour activity cycle – sleep, sedentary behavior (SED), light intensity physical activity (LPA) and moderate-to-vigorous physical activity (MVPA) – may now be feasible using small wearable devices.
Purpose: This study compares nine devices for accuracy in 24-hour activity measurement.
Methods: Adults (N=40, 47% male) wore nine devices for 24-hours: Actigraph GT3X+, activPAL, Fitbit One, GENEactiv, Jawbone Up, LUMOback, Nike Fuelband, Omron pedometer, and Z-Machine. Comparisons (to standards) were made for total sleep time (Z-machine), time spent in SED (activPAL), LPA (GT3x+), MVPA (GT3x+), and steps (Omron). Analysis included mean absolute percent error, equivalence testing, and Bland-Altman plots.
Results: Error rates ranged from 8.1-16.9% for sleep; 9.5-65.8% for SED; 19.7-28.0% for LPA; 51.8-92% for MVPA; and 14.1-29.9% for steps. Equivalence testing indicated only two comparisons were significantly equivalent to standards: the LUMOback for sedentary behavior and the GT3X+ for sleep. Bland-Altman plots indicated GT3X+ had the closest measurement for sleep, LUMOback for sedentary behavior, GENEactiv for LPA, Fitbit for MVPA and GT3X+ for steps.
Conclusions: Currently, no device accurately captures activity data across the entire 24-hour day, but the future of activity measurement should aim for accurate 24-hour measurement as a goal. Researchers should continue to select measurement devices based on their primary outcomes of interest.